Resonant Cavity Type Microwave Dielectrometer
Resonant Cavity Type Microwave Dielectrometer calculates dielectric properties from resonance changes in the resonant cavity. By just inserting a sample material in the cavity and following the multi-step wizard of the software, one can easily obtain desired quantities.
A sample setting of a resonant cavity for a flat panel (JIS-compliant 1641)
A sample setting of 2-10GHz resonant cavity (JIS-compliant C2565)
- Fast and accurate
- Easy to operate
There are two types of components: oscillator type and network analyzer type. The oscillator includes a microwave signal generator and a detecter, and serves as a substitute network, thereby eliminating the need of a costly network analyzer. The network analyzer type should be used with a network analyzer of your own.
an oscillator, a measurement cavity, a customized software, and accessories.
(PC is not included)
a measurement cavity, a customized software, and accessories.
(Neither PC nor network analyzer is included)
|Frequency range||1 - 18GHz|
|Frequency point||1 frequency point per a resonator|
|Measurement range||ε: 1 - 30 tanδ: 0.0001 - 0.1|
|Max. measurement error||ε:±1% tanδ: ±5%|
|Sample Shape||strip, square bar, thin film|